Air Surgical Unit
New command unit for Air Power hand-piece
Independent unit (patented), automatically supply sterile irrigation to inserts. It exclusively functional with compressed air (no electricity), allows the control of all sonic vibrating hand-pieces. Furthermore, safe for pacemaker wearers, risk-free performance in with flammable gas filled environments (operating theatre).
Operating air is supplied through a quick coupling tube to a dental unit. Sterile liquid irrigation is supplied by use of disposable soft plastic bag. Irrigation liquid is pressed by the inflatable bag-press which is included in the UNIT. The irrigating liquid flow, unlike peristaltic pumps, is constant and not intermittent.
In “Auto” position the sterile irrigation and hand-piece are automatically activated simultaneously by pedal to start.
In “On” position it allows to place or remove the infuser tube from seat.
Measures the air pressure in the bag-inflation device. When over 0.7 BAR an automatic safety valve is activated.
Bag-inflation pressure regulator
Adjustable setting knob for maximum air pressure in the bag-inflation device, to establish the flow of the irrigating liquid.
Handpiece cord connector
Screwable ring hand-piece air cord panel connection; small hole entrance and large hole exit.
Air pedal connectors command
Quick coupling for air tubes, male and female respectively, to and from the control pedal.Pre ssure gauge and handpiece regulator
Pressure gauge and hand-piece regulator
Pressure measurement gauge (left) and adjustable setting knob (right) for air powered hand-piece of up to 4.4 BAR max.
Pneumatic tube crushing device
A: Plunger in stand-by position, irrigating tube is open and liquid flows freely.
B: Activated plunger, tube is crushed and irrigating liquid flow is interrupted.